Working Principle: Electric operating bed adopts electro-hydraulic system. The main control structure consists of control switch, speed regulating valve and solenoid valve. The hydraulic power source is provided by the electro-hydraulic gear pump. The reciprocating motion of each bidirectional hydraulic cylinder is controlled and the position of arsenic is changed according to the healthy control of the operating bed, such as lifting, left-right tilting, rigorous backward tilting, Yao back lifting and moving with equal work. Yes. Make it meet the requirements of operation.
Normal maintenance and maintenance and matters needing attention
(1) Electric operating table in the long-term use process - the most important thing is to keep the operating table clean. After the operation, cut off the power supply, wash the outside of the operating bed to remove blood stains, and spray disinfectant on the dirt. Do not use strong corrosive or acidic detergent and disinfectant. Rinse with water is strictly prohibited. When washing and disinfecting the ground, the bottom wheel of the universal operating bed should be dropped and pushed to a dry place to prevent dampness in the inner part.
(2) Because the multi-functional operating bed uses hydraulic pressure, the oil tank should be checked regularly, so that the bed can be lowered to the lowest position and the remaining oil of hydraulic oil in the tank can be checked, but it should be 3/4 when the oil tank is insufficient. And change oil every two years.
Company Name:Vega Medical Equipment(Shanghai)
Contact: Laura duan
Company Add:Room 512, Lvdifangzhou Building, No. 258, North Of Yunhe Road, Fengxian District, Shanghai, China (Mainland)
Factory Add1:No.618,Jianyun Road,Meilin Village,Zhoupu Town,Pudong New Area,Shanghai,China
Factory Add2:No.688,Guangjian Road,Yangwang Industrial Zone,Fengxian District,Shanghai,China