As the operation shadowless lamp is more and more popular today, now know about the installation and debugging skills operation shadowless lamp is more and more important, the following simple Vega Medical Equipment(Shanghai) for you to tell me the shadowless lamp installation should pay attention to what the operation.
First.surgical light mechanical installation inspection: to check whether there are mechanical installation mistakes, whether there are less screws or screws. The card spring has no card, all the decorative cover plate has been covered.
Second, circuit inspection: first check whether the system has short circuit or break. If all normal, can conduct electric test, at this time should check the external power supply of shadowless lamp is stable. Whether the transformer output voltage is too high or too low.
Three, balance arm adjustment: check the balance arm and lamp holder match, whether also need to adjust the balance arm of force, whether need to adjust balance arm Angle and can be adjusted through the adjustment of the lamp arm method. The screw of the lamp arm is usually adjusted at both ends of the lamp arm, but different manufacturers can observe carefully.
Four. Operation light flexibility adjustment of joints: mainly the damped screw of the adjustment joint. The standard regulation of the damping adjustment is that the force of the joint is either pushed or rotated at any position to be close to 20N or 5Nm.
Five, the joints, the adjustment and the loose rev limit switch: to facilitate the shadowless lamp installation and transportation, most of the astral lamp products in its key parts such as joints all set the lock switch or limit switch. It's all open now.
Six. Check the distance and depth of irradiation: most astral lamp's irradiation distance is 70-140cm or 60-130cm. That is, shadowless lamp works best in this distance. 7. Check illumination and color temperature: illumination and color temperature are actually an approximate concept. It's all about the intensity of exposure to the shadowless lamp. Usually we think that the ideal shadowless lamp is the light and the sunlight. That's around 4,400-4600k.
After finishing the above adjustment and examination, the operating shadowless lamp can be put into use.
Company Name:Vega Medical Equipment(Shanghai)
Contact: Laura duan
Company Add:Room 512, Lvdifangzhou Building, No. 258, North Of Yunhe Road, Fengxian District, Shanghai, China (Mainland)
Factory Add:No.618,Jianyun Road,Meilin Village,Zhoupu Town,Pudong New Area,Shanghai,China
No.688,Guangjian Road,Yangwang Industrial Zone,Fengxian District,Shanghai,China