The electric surgical bed consists of table, main body, electronic control and accessories. Table composition: it is composed of head board, back board, seat board and leg board, including head board, back board, seat board, left leg board, right leg board and waist board. This combined design not only realizes the operation of lifting and lowering of the surgical bed, tilting left and right before and after, as well as the control of the head, back, waist and leg board, but also can realize the adjustment of surgical posture only through the mesa transformation.
On the working principle of the surgical bed by hydraulic power, by the control switch, speed regulating valve and solenoid valve control structure of the main body, through the electric hydraulic gear pump with hydraulic power source, control the bidirectional hydraulic cylinder reciprocating motion, and through the handle button control operation bed for a variety of position transformation, such as lift and tilt around, pour before and after, the small of the back lifting fixed, mobile, and other functions, so as to meet the requirement of operation.
The hydraulic system is characterized by compact structure, small volume, light weight, low noise, stable control, easy installation and overall layout. The hydraulic system includes oil tank, one-way valve, solenoid valve, overflow valve, bursa type energy storage device, pressure gauge, throttle valve and oil cylinder. The two sides of the surgical bed are equipped with a close control bed controller, and various motion controls can be completed through the hand controller.
The parameters of the electric surgical bed are controlled by microprocessors. The microprocessor is connected to the hydraulic and motor actuator through the control drive part. The hydraulic and motor actuator part is used to complete various movements of the surgical bed, including front and back movements, longitudinal movements, elevations and so on. Scientific circuit design is adopted to greatly reduce the fault point of the circuit and ensure the reliability of the action.
Company Name:Vega Medical Equipment(Shanghai)
Contact: Laura duan
Company Add:Room 512, Lvdifangzhou Building, No. 258, North Of Yunhe Road, Fengxian District, Shanghai, China (Mainland)
Factory Add1:No.618,Jianyun Road,Meilin Village,Zhoupu Town,Pudong New Area,Shanghai,China
Factory Add2:No.688,Guangjian Road,Yangwang Industrial Zone,Fengxian District,Shanghai,China