With the increasing popularity of surgical shadowless lights today, it is becoming more and more important to understand the installation and debugging techniques of LED Medical Operation Surgical Light.
First, the mechanical installation inspection: Check whether there is no mechanical installation leakage, with or without little screws or screws. Whether the card spring is good or not, and whether the various decorative covers have been covered.
Second, the circuit checks: First check whether the system has a short circuit or open circuit. If everything is normal, you can conduct power test. At this time, check whether the external power supply of the shadowless lamp is stable. The transformer output voltage is too high or too low.
Third, the balance arm adjustment: to see if the balance arm and the lamp head match, whether it is necessary to adjust the force of the balance arm, whether the need to adjust the angle of the balance arm plus the need for adjustment can be adjusted by the lamp arm. Normally, the lamp arms are adjusted at the two ends of the lamp arm, but different manufacturers may have slight differences and can be carefully observed.
Fourth, the adjustment of joint flexibility: mainly to adjust the joints of the damping screw, the standard provisions of the damping adjustment of the degree of tension is in any position to push or rotate the joint force is close to 20N or 5Nm.
Fifth, the joint limit switch adjustment and loose Kai: In order to facilitate the installation and transportation of shadowless lamps, most shadowless lamp products in their key parts such as joints are set lock switch or limit switch. Can now open.
Sixth, check the irradiation distance and depth: most of the shadowless lamp irradiation distance is 70-140cm or 60-130cm. That is, the shadowless lamp works best in this distance. Seven, check the illumination and color temperature: illumination and color temperature is actually an approximate concept. It is to describe the illumination intensity of the shadowless lamp. Usually we think that the ideal shadowless light is the light and the sun. That is around 4400-4600K.
After the above several adjustments are made, the surgical shadowless lamp can be put into use.
Company Name:Vega Medical Equipment(Shanghai)
Contact: Laura duan
Company Add:Room 512, Lvdifangzhou Building, No. 258, North Of Yunhe Road, Fengxian District, Shanghai, China (Mainland)
Factory Add1:No.618,Jianyun Road,Meilin Village,Zhoupu Town,Pudong New Area,Shanghai,China
Factory Add2:No.688,Guangjian Road,Yangwang Industrial Zone,Fengxian District,Shanghai,China