The surgical shadowless design and installation position and high standard sealing handle can effectively control the number of pathogens and can be disassembled and disinfected.
1. Daily inspection: lamp working status (PRX6000 and 8000). Method: Use a piece of white paper to put it in the work area. If there is an arc shadow, change the corresponding bulb. The disinfection handle is in place. Method: Two “咔哒” when installing. Cleaning: Wash the surface with a weakly alkaline solvent (soap water); avoid using chlorine-containing lotions (injury metal) and alcohol washes (injury plastics and paint).
2. Monthly inspection: Mainly check whether the standby power system (battery) is normal. Method: Turn off the 220V power supply to see if the backup power is activated.
3. The lamp life is 1000 hours on average; for the lamp holder, it is basically replaced once a year. The premise is to use the lamp specified by the manufacturer.
4. Annual inspection: You can ask a professional manufacturer to send someone to test. Replace the aging parts.
Company Name:Vega Medical Equipment(Shanghai)
Contact: Laura duan
Company Add:Room 512, Lvdifangzhou Building, No. 258, North Of Yunhe Road, Fengxian District, Shanghai, China (Mainland)
Factory Add1:No.618,Jianyun Road,Meilin Village,Zhoupu Town,Pudong New Area,Shanghai,China
Factory Add2:No.688,Guangjian Road,Yangwang Industrial Zone,Fengxian District,Shanghai,China