As a key equipment of shadowless lamp, it needs to be wiped. As an important equipment in operating room, shadowless lamp should be maintained in daily use of traditional Chinese medicine to avoid problems during the next operation.
Daily Inspection: Commonly used inspection method of the working state of the bulb: put a piece of white paper in the working area, if there is arc shadow, change the corresponding bulb. Inspection method for sterilizing handle in place: two clicks when installing. Cleaning method: Use weak alkaline solvent (soap water) to wash the appearance; Avoid using chlorine lotion (damaging metal) and alcohol lotion (damaging plastic and paint). Monthly check: mainly check whether the standby power system is normal. Methods: cut off the 220V power supply and see if the backup power is activated. The bulb life is 1000 hours on average, and the lamp holder is basically replaced once a year. The premise is to use the bulb specified by ALM. With the continuous progress of medical and health undertakings, the requirement for medical equipment is getting higher and higher. Shadowless lamp, as an operating room equipment, has been paid more and more attention by hospitals.
In the present operating room equipment, the operation shadowless lamp is a kind of more important equipment, it can satisfy the operation lighting use.
Company Name:Vega Medical Equipment(Shanghai)
Contact: Laura duan
Company Add:Room 512, Lvdifangzhou Building, No. 258, North Of Yunhe Road, Fengxian District, Shanghai, China (Mainland)
Factory Add1:No.618,Jianyun Road,Meilin Village,Zhoupu Town,Pudong New Area,Shanghai,China
Factory Add2:No.688,Guangjian Road,Yangwang Industrial Zone,Fengxian District,Shanghai,China