As one of the necessary medical devices in the operating room, the operating table is undoubtedly an important helper for surgeons during the operation. It not only allows the patient to be placed smoothly, but also can automatically change various postures according to the needs. At the same time, Maintenance loss rate will also be accelerated, then Vega Medical Equipment(Shanghai) to introduce the daily care precautions;
Surgical bed working principle: the use of electro-hydraulic system, the control switch, speed control valve, solenoid valve composed of the main control structure through the electric hydraulic gear pump to provide hydraulic power source to control the reciprocating motion of each bi-directional hydraulic cylinder and control by operating the operating table Arsenic changes in the position such as movements, leftward and rightward, harsh dumping, Yao back to help lift,
Mobile with the same features. So as to achieve the requirements of surgical operation. Normal maintenance and care and precautions.
(1) In the long-term use of the process - the most important thing is to support the operating table clean. Turn off the power after the operation is completed, to help wash the outside of the operating table to help remove blood stains, stains can be sprayed disinfectant avoid using strong corrosive or acidic cleaners and disinfectant is also prohibited from washing with water. Flush disinfection of the ground should be the end of the operating table down to push it to the dry place to prevent moisture within the help.
(2) Because it is hydraulic should be checked frequently Fuel tank can be reduced to the lowest position of the bed See the remaining amount of hydraulic oil in the fuel tank less than 3/4 of the tank should be added to 3/4 (if marked in accordance with the label) . And every two years will have to change the oil Once the use of manufacturers designated models have extended the service life of equipment.
Basic fault repair
(1) The bed body of one operating bed can not be raised but can be lowered. The inspection found that the motor did not start up when the ascending key was pressed. Power off the handle control box, open the test button with a multimeter normal, continue to measure the connection line and found that the rise of the control line open circuit, re-welding, work properly. Since then there have been other places uncontrolled failure, the check is the control box connection fault, because the cable factory is very short, coupled with the improper use of the operator, after a few repairs in the cable more and more The shorter, the last we find a longer cable, according to the original welding again, use so far.
(2) an operating room bed, staff found on one side of the bed there is the phenomenon of oil spills, and lumbar dysfunction. Opened the operating bed on both sides of the shield, after careful examination to find the place of oil leaks found in pressing the "lumbar up" key, the oil flow from one end to both ends of the hydraulic cylinder should be caused by the tight seal of the cylinder . Power off, remove the hydraulic cylinder, open the seal is sealed because the service life of a long aging, began to slowly exuded, and later more and more gaps, so that the fuel tank to reduce the oil, leading to some malfunction. Find the cause of the malfunction, replace the new gasket, tighten and debug, add enough oil, so that the various functions of the operating bed back to normal. Requires operation.
Company Name:Vega Medical Equipment(Shanghai)
Contact: Laura duan